The change of state position is independent of the actual syringe near end of infusion point. If abnormal performance is observed, additional measures may be necessary, such as re-orienting or relocating the pump. To print the manual completely, please, download it. Also check that the handset LED lights manyal then goes off. To assess the electromagnetic environment due to fixed RF transmitters, an electromagnetic site survey should be considered.
|Published (Last):||14 March 2007|
|PDF File Size:||14.76 Mb|
|ePub File Size:||14.94 Mb|
|Price:||Free* [*Free Regsitration Required]|
Changing the clock will automatically: This pump contains static-sensitive components. To convert Kilograms of Force kgf to Newtons N multiply by 9. Remove the pin securing the leadscrew gear. CareFusion Corporation or one of its subsidiaries. Italics Used to refer to other documents or manuals. Where an item is marked as Fail, the original setting is not changed.
Select the Settings tab then configure settings as shown below. Crop pin 1 to within 1mm of the board. Wherever this symbol is shown an Update note is found. Turn the pump OFF. If it does not 7. The keypad is polled from the Control PCB. This circuit has high output voltage. Once finished, the Event Log data will be displayed on the PC screen.
If the user of the pump requires continued operation during power mains interruptions, it is recommended that the pump be powered from an uninterruptible power supply or a battery. Remove the two screws securing the motor assembly to the mounting plate. Remove the screw A securing the handset earth braid. Note 1—UT is the AC mains voltage prior to application of the test level.
Entering an Access Code Each menu and certain individual options has its own three-digit technical access code which is entered using the following procedure: Replace the alarm assembly with new alarm assembly. Audible Alarm The audible alarm is located on the power supply PCB and can be enabled either by the audible alarm drive, or the watchdog alarm signal.
An electrical shock hazard exists if the casing of the pump is opened or removed. Fit the new actuator B and secure with the two screws A. Perform upgrades by acquiring the software upgrade kits specified in spare parts listings. To reassemble the components into the new lower case, simply reverse the order of disassembly.
Check cover detect reed switch operation by running Self-Test Lay the capacitor horizontally and support by applying a small quantity of silicon sealant. O n the Transfer menu, select Capture Text. Check flexi circuit, replace if necessary. Alarm Audible alarm Check that the alarm sounds continuously. Conditions of Use The Pump should only be operated by a clinician competent in use of automated infusion pumps and post-placement management of intravenous catheters.
Control PCB Contains the main processor module which provides the control functions for almost all aspects of the pump. Linear Speed Optical Switch A slotted optical switch is mounted on the transmission carriage and is used in conjunction with a slotted grid to monitor the linear travel of the pump.
Loading doses are not available. Pump in normal mode. Disconnect the flexi circuit from the Control PCB. Select NEXT to continue to the next screen. Only use CareFusion recommended spare parts. Most Related.
Cardinal IVAC 572 Setup manual
Cardinal IVAC 571, 572 Service manual
IVAC 572 MANUAL PDF